You are here: Home> Blog News> Technical Documents
Maintenance method of PRP beauty centrifuge
Author: BLLA Published: 2012-10-29 Visits: 2437
Maintenance method ofPRP beauty centrifuge: Use water or soft detergent to clean the rotor chamber and rotor. Do not use alkaline solutions or solvents that have abrasion on the material. When the centrifuge is not in use, open the top cover and keep the rotor chamber dry. Avoid motor bearing wear; use rags or tweezers to remove debris from the rotor chamber. Maintenance methods of PRP centrifuge accessories: The centrifuge should be kept as far away from other electrical equipment as possible, with good grounding measures and periodic inspections; the centrifuge and rotor must not be exposed to high intensity UV radiation or long-term heat.Use a neutral detergent when cleaning; if necessary, the rotor can be replaced.After reinstalling, tighten the head screws. If the centrifuge tube shows color changes, deformation, leakage, etc., you must stop using it. Do not screw on the tube cap when the centrifuge tube is sterilized at high temperature and high pressure to avoid deformation of the tube.See the section in this section of the user manual for the temperature tolerance of each centrifuge tube. Maintenance method of PRP centrifuge rotor pin: The rotor pin needs to be lubricated with lubricating oil frequently to ensure the centrifuge runs smoothly. Sterilization and disinfection methods of PRP centrifuge rotor cavity and accessories: all conventional disinfectants can be used.Since centrifuges and accessories are made of different materials, the compatibility of disinfectants must be considered. The operator of the PRP centrifuge should ensure that the important parts of the centrifuge are intact, mainly including: the motor is stable, the rotor and accessories are not corroded, the shaft is not deviated, the screw connection is tight, and the ground is good.